Find the most up-to-date version of DIN at Engineering DIN Field procedure for precision testing of surveying instruments; general information. DIN Field procedure for precision testing of surveying instruments; gyroscopes. standard by Deutsches Institut Fur Normung E.V.
|Country:||Turks & Caicos Islands|
|Published (Last):||7 November 2010|
|PDF File Size:||10.33 Mb|
|ePub File Size:||20.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Now, if a surveyor is using poorly defined targets in poor seeing condititions, sure, all bets are off. Submit a new text post. See below for screen shot of my data sheet of my instrument. However, because an operator normally carefully monitors the bubble and attempts to keep it centered while turning angles, the amount and direction by which the instrument is out of level become random, and hence the resulting errors tend to be random.
The optical plummet can be checked by rotating the knurled eyepiece until the cross hairs are the thickest and blackest, refocus and check for parallax. In the real world there will be many other sources of error in the operation. Kent, I guess you dni that you are an anomaly. Here is a field procedure to determine the DIN Forum contains no unread posts Forum contains unread posts Mark all read Topic Icons: I mean that the DIN number you come up with for that instrument is specific to that one instrument.
Ensure the instrument is set properly over the point.
Accuracy relates to the quality of the result obtained when compared to the standard. If a company makes 1, 2, 3, 5, and 7″ guns, it’s not because they manufactured them differently, it’s because the 7″ gun didn’t meet the 1″ spec.
RPLS Today is a peer-to-peer platform providing news, 187223, and community-based support for Ddin Surveyors and Geomatics Professionals all over the world. Do you know how your tilt compensator works? My wording was confusing. We think you have liked this presentation.
Then again, at that point you’re really splitting hairs, but I’d be curious as to how different the numbers are. These are the steps I did on a summary for a lecture. Now that I have purchased this instrument and I have taken 20 repeated angle observations of the same angle, then am I correct in thinking that if I calculated the mean of my 20 angle observations, all 20 of my angle observations will be within 2″ of the mean – have I interpreted this correctly?
The error in an angle is given by the equation:. Compute the residual for each set and each target and the mean residual. Even a robot has to be set up by a person over the point coupled with the rod being plumb. If you wish to download it, please recommend it to your friends in any social system. Forums Recent Posts Login Register. Did you by chance cover the steps for zenith angle, levels, or EDM in your lecture?
Recheck the instrument level If the instrument has moved, check the angle just measured. Welcome to Reddit, the front page of the internet.
We’ve established a quality community of surveying friends, mentors and enthusiasts, where we can all hang out, chat and have some fun sharing our experiences and helping each other.
The first one is for the “testing theodolites” and the second is for levels. What is the estimated error in the angle due to pointing and reading? If the object appears to move, parallax exists. If you look at the precision specs of a Trimble instrument, the angular precisions are usually given using DIN standards.
Amount and quality of data acquired GPS signal strength and continuity Ionosphere and troposphere conditions Station site stability, obstructions, and multipath. I ain’t ever got 187233 where, but I got there by hard work. GCP coordinate system for processing Our newest member: Space Weather influence on satellite based navigation and precise positioning R.
Yes, the most fundamental and easy way to upgrade angular accuracy is just to 188723 well-designed targets that are up to the task and use methods that actually center them over the ground marks they are nominally occupying.
Land Surveying Community Forums
I mean really know all the math behind it? Please Login or Register. I mean, if you get it, pass it on.
I want the full DIN standard to see if there’s any other useful stuff that I should know about these precisions. On hillsides- one leg uphill and two legs downhill If the ground in soft or muddy, drive hubs in the ground to support the tripod legs. If you look at the title of DIN its: About project SlidePlayer Terms of Service. Want to add to the discussion?
It may have been replaced by Djn Forum contains no unread posts Forum contains unread posts Mark all read. August 3, 9: Sad fact, too many surveyors use poorly defined targets. Before hitting submit, check that your content doesn’t violate the rules below: Factor in real world errors such as sighting and targeting errors, we can show that our measurements, unless being operated in a laboratory setting, will never achieve the numbers stated by the manufacturer.
I have the 12th edition at hand. So negliglible improvements resulted.